Does size matter? Separations on guard columns for fast sample analysis applied to bioenergy research
© Bauer and Ibáñez; licensee BioMed Central. 2015
Received: 18 December 2014
Accepted: 1 May 2015
Published: 28 May 2015
Increasing sample throughput is needed when large numbers of samples have to be processed. In chromatography, one strategy is to reduce column length for decreased analysis time. Therefore, the feasibility of analyzing samples simply on a guard column was explored using refractive index and ultraviolet detection. Results from the guard columns were compared to the analyses using the standard 300 mm Aminex HPX-87H column which is widely applied to the analysis of samples from many biotechnology- and bioenergy-related experiments such as biomass conversions or fermentations.
The 50 mm Rezex RFQ Fast Acid H+ guard column was able to separate the most common fermentation products (ethanol, acetone, iso- and n-butanol) and promising precursors (furfural and 5-hydroxymethylfurfural) of biofuels and value-added chemicals. Compound profiles in fermentation samples were analyzed with similar accuracy compared to results using the 300 mm column. However, separation of glucose and xylose was not achieved. Nevertheless, it was possible to monitor the consumption of one of the two sugars during fermentation if the other one was absent or remained constant over the course of the experiment. If correct peak integration and interference subtraction was applied, concentration profiles from enzymatic digestibility experiments and even more complex samples (e.g. acetone-butanol-ethanol (ABE) fermentation) were reliably obtained. With the 50 mm guard column, samples were analyzed up to ten-times faster compared to the 300 mm column. A further decrease in analysis time was achieved by using the 30 mm Micro Guard Cation H guard column. This column is especially suitable for the rapid analysis of compounds with long elution times on the standard 300 mm column, such as biofuel-related alcohols (e.g., n-butanol, n-hexanol) and furan- and tetrahydrofuran-type molecules.
Applied to a suitable set of samples, separations on a guard column can give rapid and sufficiently accurate information on compound changes over the course of an experiment. Therefore, it is an inexpensive and ideal tool for processing a large amount of samples, such as in screening or discovery experiments, where detecting relative changes is often sufficient to identify promising candidates for further analysis.
KeywordsIon exclusion chromatography Aminex HPX-87H Rezex RFQ Micro Guard Cation H Enzymatic digestibility assay Glucose Ethanol Butanol Furans Fermentation
A popular liquid chromatography column used for the analysis of biomass conversion and fermentation products is based on a polymer matrix of polystyrene-divinylbenzene (e.g. Bio-Rad Aminex HPX-87H, Phenomenex Rezex ROA and RFQ, Shodex SH1821, Sigma-Aldrich Supelcogel C610H and others) [1–6]. It is operated in isocratic elution mode enabling the connection to a refractive index detector (RID) which provides universal compound detection. This set up allows for analysis of a wide range of compounds. With a standard 300 mm length, this type of column provides baseline separation of the main cell wall sugars (glucose, xylose and arabinose) of many lignocellulosic biomass feedstocks. This column is also used for the analysis of organic acids, alcohols (e.g., ethanol, n-butanol) and sugar degradation products (e.g., 5-hydroxymethylfurfural, furfural) [7–10]. It only requires acidified water as mobile phase and only minimal sample preparation.
Typical HPLC methods aim to separate complex mixtures in order to individually detect isolated compounds – particularly when a non-specific detector like RID is used. In our experience from supporting researchers in an organization comprising more than 500 scientists dedicated to bioenergy research, many analytical questions can be reduced to identifying the change in a few compounds over the course of the experiment. Furthermore, reducing analysis time is often more important than obtaining precise absolute quantification data. This is especially true for the results from screening experiments involving a larger amount of samples. The detection of relative changes is often sufficient to identify promising candidates for further analysis.
Fast methods targeting individual compounds in complex mixtures have previously been established and include, for example, gas chromatography  spectroscopy , or enzymatic assays, such as for sugars, sugar alcohols, organic acids, and ethanol [13–17]. Several of these target analytes can even be measured in an automated mode using a biochemistry analyzer (YSI) and results are obtained in minutes . However, all such methods either require costly equipment or individual analysis kits for every single analyte. Therefore, it is tempting to use existing analytical instrumentation (HPLC) and reduce analysis time for higher sample throughput. In this respect, Scarlata and Hyman  have successfully shown that reducing the column length to 100 mm (plus a 30 mm guard column) can reduce the analysis time by a factor of five compared to the 300 mm column, but still provide adequate accurate results from a biomass compositional analysis (glucose, xylose, arabinose, acetate, 5-hydroxymethylfurfural, furfural). In our studies, we explored a further reduction of the analysis time by using only a guard column as the simplest and most basic column available. For this purpose, samples generated by various research groups in our institute were analyzed on the standard 300 mm Aminex HPX-87H column and compared to results obtained from an analysis on a guard column (50 mm Rezex RFQ Fast Acid H+ as well as 30 mm Micro Guard Cation H).
Results and discussion
A short column length is often used for faster separation especially in conjunction with mass spectrometric detection. These shorter columns have usually smaller particle sizes (3.5 μm and lower) which greatly improve resolution. However, columns like the Aminex HPX-87H operate in a mixture of size- and ion-exchange/exclusion. They consist of a resin with a rather large particle size (8–9 μm). As a result, a reduction in column length significantly lowers column efficiency. For example, a reduction of the column length to 50 mm results in practically unresolved glucose and xylose (see peak 1 and 2 in Fig. 6a). These types of columns can therefore not be directly compared to other columns. Despite the lower separation efficiency, the shorter columns can still provide adequate chromatographic resolution when used with appropriate samples in which compounds co-eluting with the target analytes are either not present or their concentration is sufficiently low so they do not significantly interfere with quantification.
Calibration results for each compound
Linear quantification range
0.01 − 50 mg/mL
0.01 − 50 mg/mL
UV 265 nm
0.01 − 50 mg/mL
UV 285 nm
0.02 − 50 mg/mL
0.005 − 50 mg/mL
0.005 − 50 mg/mL
0.005 − 50 mg/mL
UV 280 nm
0.00005 − 0.25 mg/mL
0.005 − 50 mg/mL
UV 280 nm
0.00005 − 0.25 mg/mL
0.005 − 20 mg/mL
0.005 − 20 mg/mL
Since 80 °C is close to the maximum operating temperature of the column (85 °C), we chose a column temperature of 55 °C for most of the following experiments. This is good balance between fast analysis time and column life time.
Application to analysis of n-butanol and iso-butanol production during fermentations
The analysis of n- and iso-butanol was performed with a 4 min total run time and this is about 9–10 times faster compared to the standard 300 mm column.
Application to analysis of 5-hydroxymethylfurfural (5-HMF) and furfural
Application to analysis of ethanol production and xylose and glucose consumption during fermentations
The consumption of xylose and production of ethanol of a modified Saccharomyces cerevisiae strain analyzed with the 50 mm guard column and 300 mm column is shown in Fig. 5b and a representative chromatogram is shown in Fig. 5a. The xylose amounts measured with the 50 mm guard column were in the range of 100.5-105.4 % and the ethanol amounts were in the range of 97.7-101.9 % (except for the two lower concentrations) of the values obtained with the 300 mm column. The two lower concentrations of ethanol showed 255.6 % (0.23 vs. 0.09 mg/mL) and 115.6 % (0.89 vs. 0.77 mg/mL) deviation most likely due to interfering co-eluting compounds at these lower concentrations. Overall, the consumption/production profile measured with the 50 mm guard column is very similar to the one obtained with the 300 mm column. However, the total run time using the 50 mm guard column was only 3 min and ethanol elution was completed at 2.5 min. This is about 10 times faster compared to the 300 mm column.
In another experiment, S. cerevisiae SA-1 was grown in a medium supplemented with 25 % hydrolysate 3 that was either pH adjusted or detoxified by membrane pervaporation which reduced the concentration of toxic compounds such as furfural, formic and acetic acid . The hydrolysate was obtained by a dilute acid and heat pretreatment of Miscanthus biomass in order to release hemicellulosic sugars and improve cellulose digestibility. During this process, the harsh conditions lead to sugar and lignin degradation resulting in the production of a fleet of fermentation inhibitors like furans, phenolics and organic acids which could also interfere with the analysis [22, 23]. Representative chromatograms of the sample with pH-adjusted hydrolysate at time points 42 h, 50 h and 67 h analyzed with the 50 mm guard column are shown in Fig. 6a-c. A decrease of glucose (peak 1) and increase of ethanol (peak 3) can be observed over the time course whereas the remaining chromatogram is mostly the same. The inset in Fig. 6a shows the close elution of glucose (peak 1) and xylose (peak 2) resulting in a peak with a shoulder making accurate integration difficult. Fig. 6d and 6e show the measured glucose and ethanol concentrations over the course of the experiments with the membrane-detoxified hydrolysate and the pH-adjusted hydrolysate, respectively. The values for “glucose uncorrected” were obtained when the attempt was made to integrate glucose separately from the closely-eluting xylose peak (Fig. 6a inset). The determined “glucose uncorrected” concentrations for experiments with the pH-adjusted and with the detoxified hydrolysate were 91.0-93.6 % and 91.6-93.0 %, respectively, of the concentrations obtained when analyzed with the 300 mm column. This indicates that xylose interfered with a correct glucose area determination. The ethanol concentrations were almost identical with 99.4-100.6 % and 99.5-100.2 %, respectively, compared to the results of the 300 mm column. When glucose and xylose were integrated together as one peak and the area of the xylose peak (peak 2, Fig. 6c) of the 67 h time point was subtracted (where glucose concentration was close to 0), “glucose corrected” values were obtained which were closer to the results of the 300 mm column (94.8-100.6 % and 99.5-99.9 %, respectively). Even without glucose correction, the effect of the toxic hydrolysate on delaying the glucose fermentation and production of ethanol until around 42 h into the fermentation was clearly observed with the 50 mm column (Fig. 6e).
Application to analysis of glucose release from enzymatic digestibility assays of pretreated biomass
Using acetate buffer, only sorbitol (peak 3) derived from the commercial enzyme mixture interfered with glucose on the 50 mm guard column (Fig. 7c) because acetate (acetic acid, peak 4) eluted later at 1.75 min. A “split-peak” integration of glucose (peak 2) and sorbitol (peak 3) is sufficient to obtain glucan conversions within 96.2-99.8 % of the ones resulting from using the 300 mm column. The 72 h conversion was 77.5 % (50 mm guard column) vs 78.5 % (300 mm column) and the run time was again about 9 times faster (1.8 min vs 16.5 min) (Fig. 7e).
It is noteworthy to mention that the acetate buffered samples can also be run with an analysis time of only 1.4 min as the citrate buffered samples. The acetate peak (peak 4) will then be eluted within the first 0.4 min of the next sample (not interfering with the glucose peak).
Furthermore, the sample analysis time of 1.4 min is fast enough to compete with other higher-throughput glucose methods like using a YSI bioanalyzer for glucose determination. With the “overlap injection” function of the autosampler, which allows rapid injection of the next sample after a run is completed, about 1000 samples (> two 384-well plates) can be processed per day.
Application to analysis of ethanol, acetone and n-butanol production during acetone-butanol-ethanol (ABE) fermentation
In the 210 nm UV trace of the same sample (Fig. 8b) a compound (peak 3) is visible that is interfering with the ethanol determination using RI detection (note that in the UV trace a compound had a slightly earlier retention time since the RI detector is in line after the UV detector). However, with the 300 mm column, peak 3 and ethanol can be separated (data not shown). If ethanol was determined only by RI detection, concentrations were overestimated and the values ranged 117.5-149.3 % (average 126.9 % +/− 8.4 %) of the values obtained with the 300 mm column (Fig. 8g). The interfering compound was present early in the fermentation process when ethanol production was zero or very minimal. Therefore, it should be possible to measure the area of peak 3 with 210 nm UV detection and correlate it to the area obtained with RI detection. In this way, the interference of peak 3 in RID mode should be reduced by measuring the sum of the area (ethanol + peak 3) with RID and subtracting the theoretical “RID area contribution” of peak 3 calculated from its 210 nm UV area. However, we noticed that the low intensity of peak 3 in the earlier time points made correlation of UV and RID signal less accurate. We therefore used an empirical method to determine the UV/RID correlation by adjusting the correlation factor so the corrected ethanol concentration from the 50 mm guard column best matched the 300 mm column results. This correction factor was applied to the UV-to-RID conversion of peak 3 at all time-points of the fermentation. “Ethanol corrected” values of average 102.8 % +/− 4.8 % (range 94.6-110.8 %) of the 300 mm values were obtained (Fig. 8g), excluding the initial 4–20 h timepoints.
Acetone (peak 5) is more selectively analyzed at a wavelength of 265 nm (absorbance maximum) but in the ABE fermentation sample an interfering peak (peak 4) was present (Fig. 8c). By applying a higher wavelength (285 nm) the area of the interfering peak was greatly reduced (Fig. 8d). Interestingly, despite the interference, even at 265 nm good analysis data (average 96.7 % +/− 5.1 %, range 85.5-105.8 % of the values using the 300 mm column, Fig. 8f) was obtained. The results from the 285 nm detection were similar (average 99.2 % +/− 5.6 %, range 87.3-106.5 % of the value using the 300 mm column, Fig. 8f), excluding the 112 h time point (113.6 %).
The increase and decrease of the profiles of ethanol, acetone and n-butanol during the course of the fermentation was caused by an applied membrane pervaporation which removed these components from the fermentation medium. During the first 50 h, the microbial production outpaced the membrane removal until the accumulated solvents became toxic to the microorganism and stopped further secretion. At this point, a decline in concentration was observed due to removal by membrane pervaporation (50–90 h). Since the concentration of the solvents was eventually reduced to a sub-toxic level, the cells resumed production of acetone, n-butanol and ethanol later in the experiment (90–112 h). Overall, analysis on the 50 mm guard column was able to monitor these changes with sufficient accuracy and with a sample analysis time of only 5 min compared to about 45 min on the 300 mm column.
Application of the Bio-Rad Micro Guard Cation H guard column
Applied to the analysis of 5-HMF and furfural (Fig. 9b), the analysis time (2.5-3 min) was further reduced by more than half compared to the 50 mm guard column (6–6.5 min, Fig. 4a). Furfural concentrations were slightly higher compared to the 300 mm column: 100.6 % (1.55 vs 1.54 mg/mL), 102.7 % (1.93 vs 1.88 mg/mL) and 109.7 % (2.60 vs 2.37 mg/mL) for hydrolysate 1–3, respectively (Fig. 4b). The concentration for 5-HMF was lower for hydrolysate 1 (64.7 %, 0.11 vs 0.17 mg/mL) and hydrolysate 3 (78.4 %, 0.69 vs 0.88 mg/mL) but identical for hydrolysate 2 (1.40 mg/mL).
An interesting application of this guard column is the analysis of some promising liquid fuel molecules and their precursors [3, 27]. Most of these types of molecules are usually analyzed by gas chromatography. If analyzed by liquid chromatography on the 300 mm column, the mobile phase has to be modified with a higher concentration of an organic solvent, e.g. acetonitrile, in order to reduce their long retention time . Instead of changing the mobile phase composition, we have explored the reduction of the retention time by using the 30 mm guard column. Fig. 9c shows the nearly baseline separation of a standard mixture of levulinic acid, 2,5-hexanedione, 5-HMF, 2,5-dimethyltetrahydrofuran, furfural, methylfuran, 5-methylfurfural and 2,5-dimethylfuran, eluting within only 4 min. Since the latter three compounds elute after furfural their elution times are in the range of 1–2 h on the 300 mm column (data not shown). Interestingly, despite the structural similarity, especially of the furans, the 30 mm guard column provided sufficient separation efficiency for these molecules. When the flow rate on the 30 mm guard column was increased to 1 mL/min a further reduction of the analysis time was achieved. At this flow rate, other compounds eluting within the first 25 min on the 300 mm column now eluted in less than 1 min on the 30 mm guard column (data not shown). For example, the elution of n-hexanol was at 2.3 min at a flow rate of 1 mL/min (Fig. 9d) compared to 96 min on the standard 300 mm column (data not shown). In general, the 30 mm guard column operated at 1.0 ml/min elutes compounds about 4 times faster than the 50 mm guard column (data not shown). Examples are, e.g., 2,5-dimethylfuran (2.3 vs 10.2 min), n-hexanol (2.2 vs 9.3 min) and 2,5-dimethyltetrahydrofuran (1.1 vs 4.3 min). The 30 mm guard column can therefore be an ideal tool for the rapid analysis of selected late-eluting compounds. We therefore encourage researchers to explore this guard column for the analysis of compounds that seem to be “invisible” to the RID when injected on the 300 mm column. This invisibility is most likely due to the effect that compounds remain on the column for a long time and are eluted as broad peaks appearing more like a “baseline-drift”, which makes peak detection and compound quantification difficult or impossible.
We have used examples from various experiments to show that a 50 mm guard column can be successfully used for the analysis of samples from many bioenergy- and biotechnology-relevant experimental setups. With this column, samples can be analyzed up to ten-times faster compared to the standard 300 mm column with very comparable results. Therefore, it is an ideal tool for processing a large amount of samples, such as in screening or discovery experiments. By applying correct peak integration and interference subtraction, concentration profiles from even more complex samples are reliably obtained. A further decrease of the analysis time was achieved by using a 30 mm guard column that has been shown to be especially suitable for the rapid analysis of compounds with long elution times on the standard 300 mm column, including biofuel-related alcohols (e.g. n-butanol, n-hexanol) and furan- and tetrahydrofuran-type molecules. Owing to the universal applicability and the ease of use as well the numerous citations in the literature, we suspect that almost all labs involved in the fields of biotechnology or biomass conversion technology are equipped with an HPLC-UV/RID instrument and should therefore be able to easily apply the methods presented here. Besides the faster analysis time, guard columns only cost a fraction of their counterparts they were designed to protect. We therefore encourage researchers to explore the feasibility of applying the separation on a guard column to their experimental setup in order to significantly reduce analysis time.
Samples were either filtered (0.45 μm) or centrifuged (10 min, 14,000 g) and analyzed using a 1200 series high-pressure liquid chromatography system (Agilent Technologies, Santa Clara, CA, USA) consisting of an autosampler with tray cooling, binary pump, degasser, thermostated column compartment, diode array detector (DAD) and refractive index detector (RI) connected in series. The supernatant was injected onto either a 300 mm × 7.8 mm (length × inner diameter) Aminex HPX-87H (Bio-Rad, Richmond, CA, USA) column with 9 μm particle size, 8 % cross-linkage equipped with a 30 × 4.6 mm micro-guard Cation H guard column cartridge (Bio-Rad, Richmond, CA) or onto a 50 × 7.8 mm Rezex™ RFQ-Fast Acid H+ guard column (Phenomenex, Torrance, CA, USA) with 8 μm particle size, 8 % cross-linkage or onto a 30 × 4.6 mm micro-guard Cation H guard column cartridge (Bio Rad, Richmond, CA, USA). Compounds were eluted either at 30, 55 or 80 °C at a flow rate of either 0.6 or 1.0 mL using a mobile phase of 5 mM sulfuric acid.
DH1, DH1ΔacrB and BW25113 E. coli strains were inoculated at an OD600 of 0.35-0.45 in 250 mL screw-capped baffled Erlenmeyer flasks in TB (terrific broth) medium (1 L contains 12 g casein peptone, 24 g yeast extract, 4 mL glycerol, 2.2 g potassium phosphate monobasic, 9.4 g potassium phosphate dibasic, Fisher Scientific, Pittburgh, PA, USA) and grown anaerobically at 30 °C and 215 rpm for up to 72 h. DH1ΔacrB strain was constructed by removing the efflux pump subunit acrB from the chromosome according to established procedures .
sFAB5441 is E. coli Dh5αZ1 strain (with LacI and TetR on chromosome) expressing iso-butanol pathway genes. The plasmids for these pathways (pSA55 and pSA69) were obtained from James Liao . Strain sFAB5692 has the same 2 plasmid as sFAB5441 but the genetic background of the strain is AL329 with several deletions (adhE; frdB/frdC; fnr; IdhA; ptA; pflB, adhP, eutG, yiaY, yjgB, betA, fucO, eutE) to improve pyruvate production and iso-butanol yield .
All strains were grown aerobically in shake flasks in LB medium and then inoculated in M9 media containing 5 g/L yeast extract, ~50 g/L glucose, and 1000-fold dilution of A5 trace metal mix (2.86 g H3BO3, 1.81 g MnCl2x4H2O, 0.222 g ZnSO4x7H2O, 0.39 g Na2MoO4x2H2O, 0.079 g CuSO4x5H2O, 49.4 mg Co(NO3)2x6H2O per liter water) at 37 °C and grown to an OD600 of 0.6-0.8. Cells were then induced with isopropyl-β-D-thiogalactopyranoside (IPTG) and then grown at 30 °C .
Pretreated biomass and hydrolysates
A two-step dilute acid pilot plant pretreatment of Miscanthus X giganteus (around 1 inch size) was performed in a pilot plant at Andritz, Glens Falls, NY. In the first step, the biomass was heat-pretreated with 0.5 % (w/w) sulfuric acid at 12.5 % (w/w) solid loading applying a rapid steam-driven heating ramp to 158 °C and a holding time of 20 min. The liquid was removed from the solids/pretreated biomass by squeezing out in a hydraulic press and was referred to as “hydrolysate 1”. In the second step, the pretreated biomass was again heat-pretreated with 1 % (w/w) sulfuric acid at 12.5 % (w/w) solid loading applying a rapid steam-driven heating ramp to 180 °C and a holding time of 4 min. The pressed out liquid from this second stage was referred to as “hydrolysate 2”.
Another pretreatment was performed in a pilot plant of the National Renewable Energy Lab (NREL), Golden, CO. Miscanthus X giganteus (around 1 inch size) was heat-pretreated with 1.5 % (w/w) sulfuric acid at 25 % (w/w) solid loading applying a rapid steam-driven heating ramp to 190 °C and a holding time of 1 min and subsequent rapid pressure release. The pretreated biomass was separated from the mixture by centrifugation and referred to as “pretreated Miscanthus biomass”. The obtained liquid phase was referred to as “hydrolysate 3”.
Xylose to ethanol fermentation
Saccharomyces cerevisiae SA-1-X123 (Brazilian industrial strain) including the X123 cassette for xylose fermentation was grown from an OD600 of 1 in yeast-peptone-xylose (YPX) medium containing 40 g/L xylose aerobically at 30 °C and 100 rpm for 62 h [31, 32].
Glucose to ethanol fermentation
“Hydrolysate 3” was either adjusted to pH 5.5 with KOH, then centrifuged and filter sterilized or detoxified by 24 h pervaporation as described previously . Saccharomyces cerevisiae SA-1 was provided by the Yeast Biochemistry and Technology Laboratory, Biological Science Department, Luiz de Queiroz College of Agriculture, University of Sao Paulo, Brazil and was grown at 30 °C at 200 rpm in 10 mL of synthetic complete media (SC-80). SC-80 contains 80 g/L glucose, 2 g/L dropout mix (US Biological, Salem, MA, USA), 6.7 g/L yeast nitrogen base (Becton Dickinson, Franklin Lakes, NJ, USA), 19.5 g/L 2-ethanesulfonic acid (MES) buffer, and a small amount of KOH to adjust the pH to 5.5. After overnight growth, cells were harvested by centrifugation. Fermentation was performed in 25 mL Hungate bottles under anaerobic conditions. The fermentation broth contained either 25 % (v/v) of “hydrolysate 3” or pervaporation-detoxified “hydrolysate 3” (with added water to match the amount removed by pervaporation), with the addition of the components of SC-80 to match SC-80 levels, and harvested SA-1 yeast cells to obtain an initial OD600 of 0.3. The fermentation was performed at 34 °C and 200 rpm for 67 h.
Enzymatic digestibility assay
“Pretreated Miscanthus biomass” was extensively washed with de-ionized water followed by consecutive centrifugation steps until pH of decanted water reached pH 5. Excess water was then removed from the biomass by manually squeezing the biomass between paper towels. An equivalent of biomass containing 1 g of cellulose was used in 25 mL liquid volume digestion reactions (considering water from biomass) in 50 mL Falcon screw cap tubes. Final digestion reactions contained either 0.05 M sodium citrate buffer (pH 4.8) or 0.05 M sodium acetate buffer (pH 4.8) and 30 FPU cellulase from Trichoderma reesei ATC 26291 (Sigma-Aldrich, St. Louis, MO, USA), 20 U Novozymes 188 (Sigma Aldrich, St. Louis, MO, USA) and 0.02 % (w/v) sodium azide. The digestion reactions were incubated at 50 °C and 200 rpm for 72 h.
ABE fermentation and pervaporation
Fermentations were carried out at 37 °C and 200 rpm in 3-L bioreactors (Bioengineering AG, Switzerland) with a 2 L working volume [33, 34]. Seed culture of Clostridium acetobutylicum ATCC824 (purchased from the American Type Culture Collection, Manassas, VA, USA) in clostridia growth medium (CGM, 100 mL) was prepared in a 150 mL anaerobic serum bottle at 37 °C until OD600 reached 2.0 and 60 mL of the seed culture was used to inoculate the bioreactor. The bioreactor automatically kept the pH ≥ 5.0 during the fermentation, using a 5 M KOH solution. Nitrogen gas was inserted into the bioreactor at a rate of 200 mL/min to maintain an anaerobic environment. Losses of volatiles, through the gas exhaust port, were minimized by using a cooling condenser attached to a RTE7 water bath (Thermo Fisher-Scientific, Sunnyvale, CA, USA) kept at 4 °C. The system was attached to pervaporation laboratory bench test unit built by Sulzer Chemtech, Neunkirchen, Germany.
After 18 h, pervaporation started with a polystyrene-block-polydimethylsiloxane-block-polystyrene (SDS) block copolymer membrane (37 cm2, 2 μm thickness (on support)) . The fermentation broth of the bioreactor was continuously passed over the membrane and back into the bioreactor by using a peristaltic pump (model # 7553–70, Cole-Parmer, Vernon, IL, USA,). ABE fermentation by C. acetobutylicum occurs in two steps: an acidogenesis phase wherein the microbes mainly produce acetic acid and butyric acid, followed by a solventogenesis phase wherein the microbes mainly produce ABE.
This work was funded by the Energy Biosciences Institute. “Hydrolysate 3” and “pretreated Miscanthus biomass” was provided by the National Renewable Energy Laboratory, 1617 Cole Blvd., Golden, CO 80401, a national laboratory of the U.S. Department of Energy managed by the Alliance for Sustainable Energy, LLC for the U.S. Department of Energy under Contract Number DE-AC36-08GO28308. The authors thank Niwen Kong, Vivek Mutalik, Thalita Peixoto Basso, Jeffrey Skerker, Zach Baer, Jon Kuchenreuther, and Eric Sacia for providing samples as well as information on sample generation and on the experimental setup. The authors also thank Mara Bryan for valuable comments and suggestions improving the manuscript. Publication made possible in part by support from the Berkeley Research Impact Initative (BRII) sponsored by the UC Berkeley Library.
- Erdei B, Barta Z, Sipos B, Réczey K, Galbe M, Zacchi G. Ethanol production from mixtures of wheat straw and wheat meal. Biotechnol Biofuels. 2010;3:16.View ArticleGoogle Scholar
- Saha BC, Iten LB, Cotta MA, Wu YW. Dilute acid pretreatment, enzymatic saccharification and fermentation of wheat straw to ethanol. Process Biochem. 2005;40:3693–700.View ArticleGoogle Scholar
- Román-Leshkov Y, Barrett CJ, Liu ZY, Dumesic JA. Production of dimethylfuran for liquid fuels from biomass-derived carbohydrates. Nature. 2007;447:982–6.View ArticleGoogle Scholar
- Brandt A, Ray MJ, To TQ, Leak DJ, Murphy RJ, Welton T. Ionic liquid pretreatment of lignocellulosic biomass with ionic liquid-water mixtures. Green Chem. 2012;13:2489–99.View ArticleGoogle Scholar
- Sluiter A, Hames B, Ruiz R, Scarlata C, Sluiter J, Templeton D, Crocker D: Determination of Structural Carbohydrates and Lignin in Biomass. In Laboratory Analytical Procedure (LAP). National Renewable Energy Laboratory (NREL), Golden, CO. Revised Version 2012. http://www.nrel.gov/biomass/analytical_procedures.html. Accessed November 2014.
- Scarlata CJ, Hyman DA. Development and validation of a fast high pressure liquid chromatography method for the analysis of lignocellulosic biomass hydrolysis and fermentation products. J Chromatogr A. 2010;1217:2082–7.View ArticleGoogle Scholar
- Buday Z, Linden JC, Karim MN. Improved acetone-butanol fermentation analysis using subambient HPLC column temperature. Enzyme Microb Technol. 1990;12:24–7.View ArticleGoogle Scholar
- Kumar M, Saini S, Gayen K. Acetone-butanol-ethanol fermentation analysis using only high performance liquid chromatography. Anal Methods. 2014;6:774–81.View ArticleGoogle Scholar
- Binder JB, Blank JJ, Cefali AV, Raines RT. Synthesis of furfural from xylose and xylan. Chem Sus Chem. 2010;3:1268–72.View ArticleGoogle Scholar
- Cai CM, Nagane N, Kumar R, Wyman CE. Coupling metal halides with a co-solvent to produce furfural and 5-HMF at high yields directly from lignocellulosic biomass as an integrated biofuels strategy. Green Chem. 2014;16:3819–29.View ArticleGoogle Scholar
- Li H, Chai X-S, Deng Y, Zhan H, Fu Z. Rapid determination of ethanol in fermentation liquor by full evaporation headspace gas chromatography. J Chromatogr A. 2009;1216:169–72.View ArticleGoogle Scholar
- Mendes LS, Oliveira FCC, Suarez PAZ, Rubim JC. Determination of ethanol in fuel ethanol and beverages by Fourier transform (FT)-near infrared and FT-Raman spectrometries. Anal Chim Acta. 2003;493:219–31.View ArticleGoogle Scholar
- Vermeir S, Nicolaï BM, Jans K, Maes G, Lammertyn J. High-throughput microplate enzymatic assay for fast sugar and acid quantification in apple and tomato. J Agric Food Chem. 2007;55:3240–8.View ArticleGoogle Scholar
- Birnberg PR, Brenner ML. A one-step assay for sucrose with sucrose phosphorylase. Anal Biochem. 1984;142:556–61.View ArticleGoogle Scholar
- Kristoffersen L, Skuterud B, Larssen BR, Skurtveit S, Smith-Kielland A. Fast quantification of ethanol in whole blood specimens by the enzymatic alcohol dehydrogenase method. Optimization by experimental design. J Anal Toxicol. 2005;29:66–70.View ArticleGoogle Scholar
- Velterop JS, Vos F. A rapid an inexpensive microplate assay for the enzymatic determination of glucose, fructose, sucrose, L-malate and citrate in tomato (Lycopersicon esculentum) extracts and in orange juice. Phytochem Anal. 2011;12:299–304.View ArticleGoogle Scholar
- Raba J, Mottola HA. Glucose oxidase as an analytical reagent. Critic Rev Anal Chem. 1995;25:1–42.View ArticleGoogle Scholar
- YSI 2950 biochemistry analyzer [http://www.ysilifesciences.com/index.php?page=ysi-2950-biochemistry-analyzer]
- Wyman CE. Aqueous Pretreatment of Plant Biomass for Biological and Chemical Conversion to Fuels and Chemicals. New York: Wiley; 2013.View ArticleGoogle Scholar
- Ebert J: Furfural: Future feedstock for fuels and chemicals. Biomass Magazine [http://biomassmagazine.com/articles/1950/furfural-future-feedstock-for-fuels-and-chemicals]
- Greer DR, Basso TP, Ibanez AB, Bauer S, Skerker JM, Ozcam AE, et al. Fermentation of hydrolysate detoxified by pervaporation through block copolymer membranes. Green Chem. 2014;16:4206–13.View ArticleGoogle Scholar
- Mitchell VD, Taylor CM, Bauer S. Comprehensive analysis of monomeric phenolics in dilute acid plant hydrolysates. Bioenerg Res. 2014;7:654–69.View ArticleGoogle Scholar
- Ibanez A, Bauer S. Analytical method for the determination of organic acids in dilute acid pretreated biomass hydrolysate by liquid chromatography time-of-flight mass spectrometry. Biotechnol Biofuels. 2014;7:145.View ArticleGoogle Scholar
- Humbird D, Davis R, Tao L, Kinchin C, Hsu D, Aden A, Schoen P, Lukas J, Olthof B, Worley M, Sexton D, Dudgeon D: Process Design and Economics for Biochemical Conversion of Lignocellulosic Biomass to Ethanol: Dilute-Acid Pretreatment and Enzymatic Hydrolysis of Corn Stover. Technical Report NREL/TP-5100-47764. United States National Renewable Energy Laboratory, US Department of Energy. May 2011.Google Scholar
- Kumar P, Barrett DM, Delwiche MJ, Stroeve P. Methods for pretreatment of lignocellulosic biomass for efficient hydrolysis and biofuel production. Ind Eng Chem Rev. 2009;48:3713–29.View ArticleGoogle Scholar
- Selig M, Weiss N, Ji Y: Enzymatic Saccharification of Lignocellulosic Biomass. In Laboratory Analytical Procedure (LAP). National Renewable Energy Laboratory (NREL), Golden, CO. 2008. http://www.nrel.gov/biomass/analytical_procedures.html. Accessed November 2014.
- Yuan J-P, Chen F. Simultaneous separation and determination of sugars, ascorbic acid and furanic compounds by HPLC-dual detection. Food Chem. 1999;64:423–7.View ArticleGoogle Scholar
- Fisher MA, Boyarskiy S, Yamada MR, Kong N, Bauer S, Tullman E. Enhancing tolerance to short-chain alcohols by engineering the Escherichia coli AcrB efflux pump to secrete the non-native substrate n-butanol. ACS Synth Biol. 2014;3:30–40.View ArticleGoogle Scholar
- Atsumi S, Hanai T, Liao JC. Non-fermentative pathways for synthesis of branched-chain higher alcohols as biofuels. Nature. 2008;451:86–90.View ArticleGoogle Scholar
- Rodriguez GM, Atsumi S. Isobutyraldehyde production from Escherichia coli by removing aldehyde reductase activity. Microb Cell Fact. 2012;11:90.View ArticleGoogle Scholar
- Kim SR, Park Y-C, Jin Y-S, Seo J-H. Strain engineering of Saccharomyces cerevisae for enhanced xylose metabolism. Biotechnol Adv. 2013;31:851–61.View ArticleGoogle Scholar
- Kim SR, Skerker JM, Kang W, Lesmana A, Wei N, Arkin AP, et al. Rational and evolutionary engineering approaches uncover a small set of genetic changes efficient for rapid xylose fermentation in Saccharomyces cerevisiae. PLoS One. 2013;8, e57048.View ArticleGoogle Scholar
- Anbarasan P, Baer ZC, Sreekumar S, Gross E, Binder JB, Blanch HW, et al. Integration of chemical catalysis with extractive fermentation to produce fuels. Nature. 2012;491:235–9.View ArticleGoogle Scholar
- Alsaker KV, Paredes C, Papoutsakis ET. Metabolite stress and tolerance in the production of biofuels and chemicals: gene-expression-based systemsanalysis of butanol, butyrate, and acetate stresses in the anaerobe Clostridium acetobutylicum. Biotechnol Bioeng. 2010;105:1131–47.Google Scholar
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.